DESCRIPTION

Insulation resistance measurements: go premium.
Sonel MIC5005 insulation resistance meter constitute a reliable tool for measuring insulation resistance up to 15 TΩ. The maximum measurement current is 1.2 mA or 3mA, which guarantees charging a highcapacity object in a short time. Each measurement, whether the programmed time expires or is manually interrupted, ends with the meter discharging the object. The device has unique algorithms that allow it to recognize the nature of the voltage present on the object, thanks to which the instrument can determine whether the voltage is the mains voltage  in which case the measurement will not be carried out  or whether it is the voltage generated by an electromagnetic field  in which case the measurement will be carried out. This makes it possible to carry out measurements in the presence of very strong electromagnetic fields where this was previously not feasible.
The MIC5005 instrument is equipped with accessories which guarantee safety during measurements with all available voltages  both test leads, probes and crocodile clips meet extremely rigorous standards (guaranteed voltage strength up to 11 kV for test leads and 5.5 kV for crocodile clips and test probes). The accessories are supplied as standard with a handy carrying case. The high resistance of the device to interference means that cables as long as 20 m can be used for measurements.
Features
 Measuring voltage in the range of 50…5000 V, 50…1000 V in steps of 10 V, 1…5 kV in steps of 25 V
 Continuous indication of measured insulation resistance and leakage current
 Automatic discharge of measured object capacitance voltage after the end of insulation resistance measurement
 Acoustic signaling of 5second intervals to facilitate capturing time characteristics.
 Adjustable measuring time  max. 99'59'’
 Metered T_{1}, T_{2} and T_{3} test times for measuring one or two absorption coefficients from the range of 1…600 s
 Measurement of coefficients: polarization (PI), absorption Ab1, Ab2, dielectric absorption ratio (DAR)
 Indication of actual test voltage during measurement
 Test current: 1.2 mA or 3 mA
 Two and threelead method of insulation resistance measurement
 Measurements with test leads up to 20 m.
 Protection against measuring live objects.
 Measurement of capacitance during the measurement of R_{ISO}
 Step voltage insulation resistance measurement (SV)
 Dielectric Discharge calculation (DD)
 Digital filters for measurements with strong interferences
Additional features
 High resistance to interferences  digital filters (10 s, 30 s, 60 s)
 Measurement of capacitance during the measurement of R_{ISO}
 Measurement of leakage current during insulation resistance testing
 DC and AC voltage measurement in the range of 0…600 V
 990 cells of memory (11880 records) with the capability of wireless data transmission to a PC (with Bluetooth) or via USB cable
 Power supply from mains or battery pack
 Backlit display
 Backlit keys
 The instruments meet the requirements of the EN 61557 standard.
TECHNICAL SPECIFICATION

Display range Resolution Accuracy 0…999 kΩ 1 kΩ ±(3% m.v. + 10 digits) 1.00…9.99 MΩ 0.01 MΩ ±(3% m.v. + 10 digits) 10.0…99.9 MΩ 0.1 MΩ ±(3% m.v. + 10 digits) 100…999 MΩ 1 MΩ ±(3% m.v. + 10 digits) 1.00…9.99 GΩ 0.01 GΩ ±(3% m.v. + 10 digits) 10.0…99.9 GΩ 0.1 GΩ ±(3% m.v. + 10 digits) 100…999 GΩ 1 GΩ ±(3.5% m.v. + 10 digits) 1.00…9.99 TΩ 0.01 TΩ ±(7.5% m.v. + 10 digits) 10.0…15.0 TΩ 0.1 TΩ ±(10% m.v. + 10 digits) U_{ISO} voltage Measuring range 250 V 500 GΩ 500 V 1 TΩ 1000 V 2.00 TΩ 2500 V 5.00 TΩ 5000 V 15.0 TΩ Display range Resolution Accuracy 0…999 nF 1 nF ±(5% m.v. + 5 digits) 1.00…49.99 μF 0.01 μF ±(5% m.v. + 5 digits)