DESCRIPTION

Measure insulation resistance up to 10 GΩ
Sonel MIC10 insulation resistance meter is a reliable device for measuring insulation resistance up to 10 GΩ. Due to its compact dimensions, ergonomic design and versatile application, this compact meter is a must for every professional working on electrical installations. It works flawlessly in areas with low voltage, such as electrical installations in single and multifamily buildings, public facilities, as well as workshops and factories.
Main features
 measurement of insulation resistance up to 10 GΩ thanks to max 1000 V measurement voltage
 designed for harsh environmental conditions  IP67 ingress protection.
 allows for testing electrical continuity  R_{CONT} 200 mA function
 checking start capacitors in motors thanks to capacity measurement function
…and much more
 measurement voltage selected from: 50, 100, 250, 500, 1000 V
 continuous reading of measured insulation resistance
 automatic discharge of the measured object’s capacitance upon completion of insulation resistance measurement
 sound signaling of fivesecond time intervals, facilitating capture of time characteristics.
 readings of actual measurement voltage during measurement
 protection against measurement of live objects
 threelead measurement
 capacitance measurement during measurement of R_{ISO}
 lowvoltage measurement of circuit continuity and resistance
 continuity test of protective conductors and equipotential bonding with current ≥200 mA flowing in two directions in compliance with EN 615574
 measurement of direct and alternating voltages within the range of 0…600 V
Application
MIC10 is an insulation resistance meter, which may be very useful for various electrical works (even the simplest), in locations with Low Voltage power supply, including electrical systems in single and multifamily buildings as well as in public buildings and in small workshops or factories. With its test voltage settings of 50 V, 100 V or 250 V the device is perfect for quick checking of the insulation condition in telecommunications systems and control cables.
Features
Test voltage settings of 500 V or 1000 V perfectly match the requirements for assessing the protection of power supply lines but also of floors and walls  PRS1 probe (optional accessory) is very useful for this purpose.
With MIC10 meter you can check whether an object is under voltage (measuring range up to 600 V), both in overhead and cable networks (measuring category of the device: CAT IV 600 V). You can verify the continuity of cables, e.g., PE connections and equipotential bonding  using the current of at least 200 mA, according to EN 615574. You can check the capacity of the startup capacitors in household appliances and drives of any type (measuring range up to 10 µF).
This makes MIC10 meter an essential tool for every service technician.
Durable housing
Handy and ergonomic housing provides protection of IP67, ensuring reliability of the meter even in the harshest environmental conditions (moisture, dust, high temperature, etc.).
TECHNICAL SPECIFICATION

Range Resolution Accuracy 0.0…999.9 kΩ 0.1 kΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 1.000…9.999 MΩ 0.001 MΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 10.00…99.99 MΩ 0.01 MΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 100.0…250.0 MΩ (for U_{n} = 50 V) 0.1 MΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 100.0…500.0 MΩ (for U_{n} = 100 V) 0.1 MΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 100.0…999.9 MΩ (for U_{n} ≥ 250 V) 0.1 MΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 1.000…2.000 GΩ (for U_{n} = 250 V) 0.001 GΩ ± (3% m.v. + 8 digits) [± (5% m.v. + 8 digits)]* 1.000…9.999 GΩ (for U_{n} ≥ 500 V) 0.001 GΩ ±(4% m.v. + 6 digits) [±(6% m.v. + 6 digits)]* 10.00…20.00 GΩ (for U_{n} ≥ 500 V)** 0.01 GΩ ±(4% m.v. + 6 digits) [±(6% m.v. + 6 digits)]* 10.000…99.99 GΩ (for U_{n} = 1000 V) 0.01 GΩ ±(4% m.v. + 6 digits) [±(6% m.v. + 6 digits)]* 10.00 GΩ (for U_{n} = 1000 V) 0.1 GΩ ±(4% m.v. + 6 digits) [±(6% m.v. + 6 digits)]* Range Resolution Accuracy 0.00…19.99 Ω 0.01 Ω ±(2% m.v. + 3 digits) 20.0…199.9 Ω 0.1 Ω ±(2% m.v. + 3 digits) 200…1999 Ω 1 Ω ±(4% m.v. + 3 digits) Range Resolution Accuracy 0…299.9 V 0.1 V ±(2% m.v. + 6 digits) 300…600 V 1 V ±(2% m.v. + 2 digits) Range Resolution Accuracy 1…999 nF 1 nF ±(5% m.v. + 10 digits) 1.00…9.99 μF 0.01 μF ±(5% m.v. + 10 digits)